# Simple pH Meter Circuit, A Low Cost Adapter for Your Digital Voltmeter

What is A pH Meter?

A pH meter is actually a precise voltmeter  that measures the generated voltage of a pH electrodes. The requirement of such meter is high input impedance and has the gain of  voltage-pH conversion. The standard pH probe generate voltage about 59 mV per pH,  so all we need is a pre-amplifier with high impedance input and with gain = 16.7 to give 1 Volt per pH. Here is the schematic diagram of the pH meter circuit:

pH Meter Circuit

This pH meter circuit use a high input impedance high gain op-amp TL081. The transformer voltage is center tapped 15V  (15V-0V-15V). Multi-turn variable resistor is recommended for the zero, slope, and reference variable resistors. Current consumption of this pH meter circuit should be very small, so the 7812 and 7912 regulator IC won’t need any heat sink.

pH Meter Calibration

To calibrate our pH meter circuit,  follow these steps:

• Null the amplifier by shorting the pH probe input to ground. Measure the voltage between output + and ground. Adjust the zero potentiometer until the voltmeter reading show zero volt (or close to zero).
• Now connect pH probe  input to a 413 mV voltage source. You can use a potentiometer to make a  simple 413 mV voltage source. After connecting the input to 413 mV source, then measure the  + output  and ground using the digital voltmeter  and you should read 7.00 volt. If not then adjust the slope potentiometer until you get the closest reading to 7.00 volts.
• Short the pH probe input to ground. Connect the digital voltmeter voltmeter between – output and + output. Now adjust reference potentiometer until the voltmeter show 7.00 volts. This corresponds to the neutral pH of 7.
• Connect the pH probe input to a pH electrodes (pH probe/sensor), now the pH meter is ready to use. The voltmeter reading in volts would be direct conversion of pH. pH value of 7 would be read as 7 volts, pH 3 as 3 volts, etc.

With low cost digital voltmeter (DVM), we can easily obtain 0.05 pH accuracy if  we follow the calibration method as described above.

This entry was posted in Test and Measurement and tagged . Bookmark the permalink.,
Possibly Related to "Simple pH Meter Circuit, A Low Cost Adapter for Your Digital Voltmeter" Circuits

Warning: ksort() expects parameter 1 to be array, object given in /home/simpleci/public_html/wp-content/plugins/yet-another-related-posts-plugin/includes.php on line 395

• Inductance Meter Adapter Circuit, Turns Your Digital Voltmeter to Inductance Meter
Measuring inductance is important in making your hand-made inductors, especially when you do-it-yourself (DIY) your coil winding. An inductance meter adapter circuit would be helpful since you can measure inductance using your general purpose digital voltmeter (DVM). This circuit cost … Continue reading →...
• Digital pH Meter Signal Conditioner with Pulse/Frequency Output
A signal conditioner for pH meter probe is required to have high input impedance. The signal conditioning of the pH meter probe is done by providing a buffer with gain, offset, and volatge to frequency converter. High input impedance requirement … Continue reading →...
• Digital pH Meter Signal Conditioner
A pH meter circuit is basically an amplifier to boost a voltage signal which is produced by a pH probe. The pH probe itself is equal to a battery cell when submersed in the measured solution, producing a weak voltage … Continue reading →...
• Capacitance Meter Circuit
Capacitance meter is one instrument that you should have in your toolbox if you are an electronics hobbyist, or if you’re a professional electronic technician. If you can’t afford to buy such instrument or you you can’t find it on … Continue reading →...
• Temperature Probe for Digital Multi Meter
This is a simple temperature measurement probe circuit. This circuit measures temperature either on a PC board or at the probe tip to which contains the temperature sensor. It can be used for debugging and troubleshooting electronic circuits. Here is … Continue reading →...